Will my data be lost when the batteries are removed?
Don’t worry—your information will be saved when you change the batteries. 1According to the ISO 15197:2013 standard, 95% of all bG results must be ±15 mg/dL of the laboratory reference when bG levels are below 100 mg/dL. At bG levels above 100 mg/dL, 95% of all bG results must be within ±15% of the laboratory reference.
How do I change the batteries?
To replace the batteries: Open the battery door on the back of the meter by pushing the tab toward the door, and pull the door up and open. Push the “hand” button to release the old batteries, then simply slide them out. With both of the plus signs facing up, slide the new batteries under the black tabs and the hand button. Replace the battery door and snap it closed. 1According to the ISO 15197:2013 standard, 95% of all bG results must be ±15 mg/dL of the laboratory reference when bG levels are below 100 mg/dL. At bG levels above 100 mg/dL, 95% of all bG results must be within ±15% of the laboratory reference.
What type of batteries are used?
The Accu-Chek Guide system uses two readily available CR2032 lithium coin batteries. 1According to the ISO 15197:2013 standard, 95% of all bG results must be ±15 mg/dL of the laboratory reference when bG levels are below 100 mg/dL. At bG levels above 100 mg/dL, 95% of all bG results must be within ±15% of the laboratory reference.
Where can I find out what an error code or screen message is telling me?
Visit your meter's Screen Messages page for a list of messages and definitions or download the User Manual for your meter.
What parts of the meter should be cleaned and disinfected?
Be sure to clean the areas around slots and openings, while being careful not to get any moisture in them. You should also wipe down the entire meter surface, including the meter display.
What can I use to clean and disinfect the meter?
70% isopropyl alcohol Mild dishwashing liquid mixed with water 10% household bleach solution (1 part bleach plus 9 parts water) made the same day
What is the difference between cleaning and disinfecting?
Cleaning is simply removing dirt from the meter, while disinfecting refers to removal of most but not all, disease-causing and other types of microorganisms.
When should the meter and lancing device be cleaned and disinfected?
Clean and disinfect your meter once a week, or any time blood gets on the meter. You should also clean the meter before allowing anyone else to handle it.
Is the time and date or memory lost when the battery is removed?
The meter includes an additional internal battery that retains the time and date for up to 72 hours. So when the battery is changed, no memory is lost.
What type of battery is used?
The meter uses 1 3-volt lithium coin cell battery (type CR 2032) found in many stores.
How do I change the battery in my Accu-Chek Performa blood glucose meter?
Make sure you have a new 3-volt, lithium battery, number CR2032, to insert. Open the battery door on the back of the meter, remove the old battery for at least 20 seconds, and press any button on the meter. Insert the new battery with the “+” sign facing out. Snap the battery door back into place. Press the on/off button, and the meter will beep. Setup and hour are flashing on the display. If the date and time are correct, press and hold the on/off button for about 4 seconds. Release the button, and you’re done.
How long will the battery last in my meter?
Your battery should last for approximately 2,000 tests.
How do I set or change the time and date on my meter?
Watch this short video to see how easy it is to set the date and time. Your meter comes with the batteries already installed and a preset time and date. You may need to change the time to match your time zone or, if you live in a state that observes daylight saving time, to reflect the time change. To change these settings: Press and release the On/Off button to turn the meter on. The flashing test strip symbol appears. NOTE: If this is the first time your meter has been turned on, it will automatically enter the set-up mode, and "Set-up" and the hour will flash on the display. Skip to Step 3 to change the time or, if the date and time are correct, press and hold the On/Off button to exit the set-up mode. To enter the set-up mode, press and hold the On/Off button for about four seconds. “Set-up” and the hour flash on the display. Press and release the arrows to decrease or increase the hour. Press and hold the arrows to scroll faster. Press and release the On/Off button to set the hour. The minutes flash. Repeat steps 3 and 4 to set the minutes, AM/PM, month, day and year. The flashing field is the one you are changing. Once you set the correct time and date, press and hold the On/Off button until the flashing test strip symbol appears.